Leaders in Flow Technology™
Still have a question? Can't find what you're looking for? Contact Customer Solutions for answers.
Customer Solutions Hours
Monday thru Friday
9:00 am - 5:00 pm US Eastern Time
Press 2 to leave a message for Tech Support after office hours.
+1 781 933 9560 Telephone
+1 888 220 2998 U.S. Toll free
Borosilicate glass and quartz PicoTips® are an improved version of the traditional nanospray ionization emitters. Their special taper makes them virtually self-filling and their specified tip opening eliminates breaking open the tips prior to use.
The proper OD for the emitter is determined by the manufacturer of your nanospray source. The Micromass (Waters) ZSpray NanoFlow uses PicoTips with a 1.0 mm OD while the ABI/Applied Biosystems/Sciex, Bruker NanoSpray, MSD Proteomics/Protana NanoES, and Thermo Scientific Nanospray Ion sources take PicoTips with a 1.2 mm OD.
Determine your target flow rate then locate the corresponding tip size on the table below. EconoTips have a 1 µm ID tip and are perfect for routine sampling with a standard single-layer conductive coating. GlassTips areavailable in 2 and 4 µm tip IDs, with or without a durable, inert, multilayer conductive coating (U.S. Patent 5,788,166) and undergo a rigorous inspection by video microscopy. Select QuartzTips for samples that are sodium sensitive or need an emitter with better resistance to high pH solvents.
|Tip Size vs. Flow Rate|
|Tip Size, ID (µm)||Flow Rate* (nL/min)|
Approximate range of flow rates over which the tips will generate a stable electrospray plume without sheath gas assistance. Actual performance may differ considerably caused by variations in experimental parameters (mobile phase composition, applied voltage, source design etc.)
Once you've determined the physical dimensions of your emitter based upon your target flow rate, how will you apply voltage to your sample; to the outside or directly to the sample?